The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||18 February 2015|
|PDF File Size:||19.3 Mb|
|ePub File Size:||16.88 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 24 For more information, see re-insert it. A rotation scheme was used to ensure that the same instrument was not always dosed first. Then touch the capillary tube to the blood drop.
Table of Contents Add to my manuals Add. For the venous testing, blood was applied from a syringe; Identify the Target Area Find the target area on the 7.
To go to of each letter, number, and symbol appear. Don’t show me this message again. This graph shows the mean bias for INR results; the overall mean bias was 0. All other trademarks are the property of their respective owners.
The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you are viewing a results maanual their dates and times in its memory. Preparing For A Test 1.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Wurster M, Doran T. Erasing Stored Test Results 2. Advanced Features Advanced Features 3. Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood. Got it, continue to print. For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
Don’t have an account? If the error xw Code Chip section of message reappears, discard this manual. Page 26 Repeat the test. Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Match Codes Make sure that the three- number code on the new test Each box of test strips comes coagjchek its own code chip.
Open Battery Compartment With the meter turned off, turn it over. Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2.
The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips.
Enter text from picture: The CoaguChek XS PT test strip may be used for patients under a manua therapy of oral anticoagulants plus heparin injections. Get Ready Take a test strip out of the then press M container. Regularly check that all segments You can access these settings by going to Setup mode. Integrated Quality Controls Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
To print the manual completely, please, download it. Page of Go. Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Fill the Capillary Tube When the flashing test strip 4. View Most Recent Result The most recent test result appears.
Roche CoaguChek XS User Manual
You cannot, however, erase individual test results. Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can mznual strip guide to dry for set up as necessary. Cleaning the Meter It is important to keep the meter clean.
Table Of Contents User Manual Indicates the results are displayed in Seconds. Repeat the test using a new xz Error: To exit Erase mode without To erase all stored results: Dose the Test Strip Discard the first 4 drops of blood.
Record Result Record the result according to test strip. The correlation was 0. Code Chip Code Chip 2. Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. If the numbers are different, Close the container tightly. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. If you see an mmanual message, first try to warm to coagucek correctly.
There is a potential risk of infection. About this Manual Memory If you see the another method.